Crops sunflower plants Soil pH Meter cone Digital pH Meter
Soil PH value plays a very important role in the growth process has grain crops, most crops in high pH soils is very difficult to survive, and therefore suitable PH value of planting the best crops vital:
1, with two-thirds into the soil when the soil compaction,not too dry.
2, when the first use of a few first inserted in the soil, remove the oxide layer on the probe.
3, press the button next to the white display is water
Usage: measurement of different soil PH value and moisture.
PH (acidity): 3-8
Accuracy: ± 0.2PH (applies to arable land and classroom experiments)
Working temperature: 5-50 ºC (Fahrenheit 41-122 ºF)
Calibration: Gently wipe to remove oxides on the metal electrode
1, remove the debris in the soil to be tested, such as topsoil, weeds, gravel, etc., if the soil is too dry or contain too much fertilizer, you need to spray some water for approximately 25-30 minutes before testing.
2, prior to testing with a damp cloth to thoroughly clean the surface of the metal electrode.
The first time you use the detector, the first electrode is inserted in the soil several times to remove surface oil so as not to affect the accuracy of PH value or humidity.
3, the detector vertically into the soil to be tested to ensure that the electrode and the surrounding soil is completely fit. After about 10 minutes, the detector will display the correct PH value and moisture.
Detector sometimes display different values, depending on the soil conditions, detector metal surface adhesion, humidity, etc. Therefore, it may take the average of several measurements as well.
4, press the meter on the side of the white or green button, the instrument will display soil moisture.
5, the detector after use, clean the electrodes with water after dry collection.
1, prior to testing to loosen the soil, be careful not to use too much force to insert excellent soil.
2. Do not insert metal part detector into the soil for too long (not more than 1 hour), to prevent electrode damage.
3, pre-test to keep the metal surface is clean and dry.
4, it can not be used to detect liquid, once the original liquid into contact with the electrical systems, fault detector occurs not covered by the warranty.
5. Do not try to take the opening tip, if you can loose tighten clockwise.
6, prohibit the use of the detector in the vicinity of magnetic objects.
7, prohibit hand holding electrode finger grease composition containing weaken current.
|Name||Portable pH Meter|
|PH measuring range||3-8 ph|
The best soil pH for Main Crops:
Crop Name Crop value
Orange, tangerine 5.0-7.0
1, if the soil is too dry or measurement points excessive fat parts, can not be measured when the soil pH, must first splash at the measurement point position, wait 28 minutes and then measured.
2, before using the measuring device must first polishing cloth, the metal parts of the absorption plate completely wipe clean to prevent influence the measured value. If not using the new metal plate surface with a protective oil, must first into the soil several times, mill net reservoir protection before use.
3, when the pH measurement point directly into the soil test, a metal plate surface must be completely buried, is the correct value obtained for about 10 minutes. Soil density, moisture and excessive fat parts may affect the measured values, it must be measured in several different positions, in order to average.
4, measured at 10 minutes after the pH is stable, then press the white button on the side, and humidity appear immediately.